A 3-DOF MR-compatible device for magic angle related in vivo experiments
Tse, Zion Tsz Ho and Elhawary, Haytham and Zivanovic, Aleksandar and Rea, Marc and Paley, Martyn and Bydder, Graeme and Davies, Brian L. and Young, Ian and Lamperth, Michael U. (2008) A 3-DOF MR-compatible device for magic angle related in vivo experiments. IEEE/ASME transactions on mechatronics, 13 (3). pp. 316-324. ISSN 1083-4435
Full text is not in this repository.
Official URL: http://dx.doi.org/10.1109/TMECH.2008.924109
This item is available in the Library Catalogue
The "magic angle" effect consists of the increase in signal intensity observed at a tendon or cartilage in a magnetic resonance image, when the tissue is oriented at an angle of approximately 55deg with respect to the main magnetic field B0. The exploitation of this phenomenon is often used to assist diagnosis of tendinous and other diseases, although practical difficulties derived from positioning target tissue at the desired orientation inside closed-bore scanners has made this exploitation hard to implement. A 3-DOF MR-compatible mechatronic system has been developed to position a variety of limbs at the magic angle inside a closed- bore scanner, actuated by a custom-developed pneumatic air motor. The system is capable of locating the desired anatomy with high accuracy, and is designed to position the target tissue at a minimal distance from the isocenter. The compatibility of the system is demonstrated, producing negligible artifacts and an insignificant reduction in signal to noise of the image. Preliminary clinical trials scanning the Achilles tendon of healthy volunteers prove the functionality of the device. An increase in signal intensity of up to 21-fold has been recorded in the tendon at the magic angle.
|Research Areas:||School of Science and Technology > Science & Technology|
|Citations on ISI Web of Science:||6|
|Deposited On:||11 Jul 2011 13:28|
|Last Modified:||30 Jul 2014 13:01|
Repository staff and depositor only: item control page
Full text downloads (NB count will be zero if no full text documents are attached to the record)
Downloads per month over the past year