Design and performance of the UCLH mark 1b 64 channel electrical impedance tomography (EIT) system, optimized for imaging brain function
Yerworth, Rebecca J. and Bayford, Richard and Cusick, G. and Conway, M. and Holder, David S. (2002) Design and performance of the UCLH mark 1b 64 channel electrical impedance tomography (EIT) system, optimized for imaging brain function. Physiological Measurement, 23 (1). pp. 149-158. ISSN 0967-3334
Full text is not in this repository.
This item is available in the Library Catalogue
The UCLH Mark 1b is a portable EIT system that can address up to 64 electrodes, which has been designed for imaging brain function with scalp electrodes. It employs a single impedance-measuring circuit and multiplexer so that electrode combinations may be addressed flexibly using software. It operates in the relatively low frequency band between 225 Hz and 77 kHz, as lower frequencies produce larger changes during brain activity, and has a videocassette-sized headbox on a lead 10 m long, connected to a base box the size of a video recorder, and notebook PC, so that recordings may be made in ambulant subjects. Its performance was assessed using a resistor–capacitor network, and two saline-filled tanks—a cylindrical Perspex one and a latex one which contained a human skull. System signal-to-noise ratio was better than 50 dB and the maximum reciprocity error less than 10% for most frequencies. The CMMR was better than 80 dB at 38 kHz and a sponge, 20 mm across, which caused a local 12% impedance increase, was correctly localized in images. This suggests that the system has adequate performance to image impedance changes of 5–50% known to occur in the brain during normal activity, epilepsy or stroke; clinical trials to image these conditions are in progress.
|Research Areas:||School of Science and Technology > Natural Sciences|
|Citations on ISI Web of Science:||27|
|Deposited On:||26 May 2009 10:07|
|Last Modified:||04 Feb 2014 07:11|
Repository staff only: item control page
Full text downloads (NB count will be zero if no full text documents are attached to the record)
Downloads per month over the past year